Pulser Features & Performance Specifications:
- Pulser will be enclosed in a standard rack mount configuration (3U rack mount configuration).
- The pulser will provide one channel output.
- Output Voltage: Positive polarity 1kV to 10kV adjustable into 50 Ohms
- Output: Single 50 Ohms coaxial e.g. HN, THT20 or other connector to be discussed and finalized with customer
- Pulse duration: Fixed duration of 10ns to 110ns, adjustable, defined as the time from 95% of PV (peak voltage) on the rising edge to 95% PV on the falling edge of the pulse. PV defined as the average value of the output voltage waveform over the pulse duration.
- Risetime: 10%PV to 90%PV ≦ 10ns.
- Fall time: 90%PV to 10%PV ≦ 10ns.
- Repetition rate: single shot or up to 5 Hz nominal operating conditions.
- External trigger: 5V logic level into 50 ohms, risetime ≦ 5ns, pulse duration 50-500ns.
- Timing jitter (with respect to external trigger input): <0.2 ns (rms)
- Timing drift (with respect to external trigger input): <+/- 0.5 ns over 24 hours (10 minutes after application of power)
- Control Interface: RS232, ethernet and front panel LCD/keypad
- Environment: Room temperature environment.
- Cooling, if required: Air cooling with inlet at front of unit and exhaust at rear.
- Primary power: 110 VAC, 60 Hz.
- Self-monitoring: switching stage health checks, load impedance, excess trigger rate
Anticipated Channel Pulse Performance
- Pulse flatness and amplitude stability: +/- 3% of PV The flatness and stability shall be measured during the time from 100% PV on the rising edge to 100% PV on the falling edge of the pulse.
- Post pulse voltage: The output voltage following the main pulse after falling to +4% will remain within +/-4% PV until the rising edge of the next pulse
- Design is tolerant of intermittent o/c, s/c and arc load faults. To avoid connector and cable damage the pulser may shut down under fault conditions.
- Service life of >50 million pulses. Self-diagnostics allow the location of failed switching devices to be identified, which can be replaced during a scheduled service as required. Note that individual main switching device failures have a minimal effect on the output amplitude.
Configure Your Product
"*" indicates required fields